Hi-Rem Polygon Post
Glass Fiber Post
Hi-Rem Polygon Post are glass fiber posts with conical-cylindrical shape.
As Overfibers post surfaces are particularly rough the silanization treatments are an optional.
Hi-Rem Polygon Post, the Overfibers’ patented fiber post, has the great advantage to be easily removable when it is necessary.
The central longitudinal axis of Hi-Rem Polygon Post is made of soft polymer macro-fiber, placing in this position doesn’t affect the resistance of the post, because the stresses are minimal (neutral axis).
Hi-Rem Polygon Post are the first “S” glass fiber posts with non-circular section present on the market. This characteristic, made possible by particular processing systems, which are part of the advanced Overfibers Know How.
They respects the morphology of the majority of the canals of single root teeth, adapting itself to the internal anatomy of the tooth canal (see photos A and B) with minimal removal of dental matter and maximum resistant section of the post.
They are recommended for very stressed teeth, such as abutments of extended fixed prostheses or in partially removable combined prosthetic anchor teeth (removable prostheses).
Particularly recommended for single root premolars, canine teeth and incisors.
- 3 Unit – Hi-Rem Polygon Post # 1
- 3 Unit – Hi-Rem Polygon Post # 2
- 3 Unit – Hi-Rem Polygon Post # 3
- 1 Polygon Drill # 1
- 1 Polygon Drill # 2
- 1 Polygon Drill # 3
Contents: 10 Unit – Hi-Rem Polygon Post # 1
Contents: 10 Unit – Hi-Rem Polygon Post # 2
Contents: 10 Unit – Hi-Rem Polygon Post # 3
Contents: 1 Polygon Drill # 1
Contents: 1 Polygon Drill # 2
Contents: 1 Polygon Drill # 3
|Highly retentive surface||Increases the post / cement* adhesion without any weakening due deep grooves!|
|“S” type glass fibers 80%||high flexure strength 1650 MPa|
|Balanced flexure E-modulus||60 GPa to better stabilize the restoration|
|Highly Translucent||provides esthetics and light transmission|
|Radiopacity||2.15 mm or 210% Al equivalent|
|Fatigue Strength||5 Millions Cycles Without Breaking|
|Sterilization in autoclave||135°C at 2.1 bar for 5 min. without any variation of their resistance|
|Easy to remove||safely with absolute certainty and without tooth damages**|
Hi-Rem Polygon Posts and Polygon Drills are available in four sizes, with color coding for a corresponding use.
|Color||Code||Maximum diameter||Minimum diameter||Point diameter||Total length||Drill|
Polygon Over Post - Cementation Techniques
Hi-Rem Post presentation
Removal of Hi-Rem Prosthetic Post
- Fiber Post Bonding Using Self-Etching and Resin Cements
S. PALLOTTINI, A. LLUKACEJ, C. MONACO, and P. BALDISSARA, University of Bologna, Bologna, Italy
- Evaluation of a Simplified Fiber Post Removal System
Nicola Scotti, DDS, Emanuele Bergantin, DDS, Mario Alovisi, DDS, Damiano Pasqualini, DDS, and Elio Berutti, MD, DDS
- Fiber Post Removal: Comparative Study Using a New Post Concept
P. BALDISSARA1, D. MELILLI2, L. CIOCCA1, and L.F. VALANDRO3, 1University of Bologna, Bologna, Italy, 2University of Palermo, Palermo, Italy, 3Federal University of Santa Maria, Santa Maria, Brazil
- Tissue Loss During Post Removal: a New Post Concept Evaluation
P. BALDISSARA1, L.F. VALANDRO2, S. BETTAZZONI1, and R. SCOTTI1, 1University of Bologna, Bologna, Italy, 2Federal University of Santa Maria, Santa Maria, Brazil
- Effect of Sterilization on Fiber Posts Fracture Strength
G. IORIZZO, S. BETAZZONI, L. CIOCCA, and P. BALDISSARA, University of Bologna, Bologna, Italy
- Microtensile bond strength of fiber post: effect of surface roughness
P. BALDISSARA1*, C. MONACO1, L.F. VALANDRO2, University of Bologna, Bologna, Italy1 and Federal University of Santa Maria, Brazil2